HX711ADC (community library)
|All Versions||0.0.3, 0.0.2, 0.0.1|
Library to interface the Avia Semiconductor HX711 24-Bit Analog-to-Digital Converter (ADC) for Weight Scales
Example Build Testing
Device OS Version:
This table is generated from an automated build. Success only indicates that the code compiled successfully.
Library Read Me
This content is provided by the library maintainer and has not been validated or approved.
An Arduino library to interface the Avia Semiconductor HX711 24-Bit Analog-to-Digital Converter (ADC) for Weight Scales.
This is my humble attempt at creating an Arduino library for this ADC: http://www.dfrobot.com/image/data/SEN0160/hx711_english.pdf
Other libraries exist, including this very good one, which I first used and which is the starting point for my library: https://github.com/aguegu/ardulibs/tree/master/hx711
Although other libraries exist, I needed a slightly different approach, so here's how my library is different than others:
It provides a tare() function, which "resets" the scale to 0. Many other implementations calculate the tare weight when the ADC is initialized only. I needed a way to be able to set the tare weight at any time. Use case: place an empty container on the scale, call tare() to reset the readings to 0, fill the container and get the weight of the content.
It provides a power_down() function, to put the ADC into a low power mode. According to the datasheet, "When PD_SCK pin changes from low to high and stays at high for longer than 60μs, HX711 enters power down mode". Use case: battery powered scales. Accordingly, there is a power_up() function to get the chip out of the low power mode.
It has a set_gain(byte gain) function that allows you to set the gain factor and select the channel. According to the datasheet, "Channel A can be programmed with a gain of 128 or 64, corresponding to a full-scale differential input voltage of ±20mV or ±40mV respectively, when a 5V supply is connected to AVDD analog power supply pin. Channel B has a fixed gain of 32.". The same function is used to select the channel A or channel B, by passing 128 or 64 for channel A, or 32 for channel B as the parameter. The default value is 128, which means "channel A with a gain factor of 128", so one can simply call set_gain(). Also, the function is called from the constructor.
The constructor has an extra parameter "gain" that allows you to set the gain factor and channel. The constructor calls the "set_gain" function mentioned above.
The "get_value" and "get_units" functions can receive an extra parameter "times", and they will return the average of multiple readings instead of a single reading.
How to Calibrate Your Scale
- Call set_scale() with no parameter.
- Call tare() with no parameter.
- Place a known weight on the scale and call get_units(10).
- Divide the result in step 3 to your known weight. You should get about the parameter you need to pass to set_scale.
- Adjust the parameter in step 4 until you get an accurate reading.
Browse Library Files